How to set up BioNet surgery monitor - BM7VET?
Modified on: Fri, 11 Oct, 2019 at 1:31 PM
IMPORTANT: We are not currently connecting any new customers as we are working to stabilize this integration feature. If you have previously connected and are already using anesthetic monitor integration, please contact our support if you need assistance.
To set up the monitor correctly, please follow the steps below.
Set the correct IP address, Gateway, Subnet Mask and Host IP for the multi-parameter portable veterinary monitor - BM7VET, (If it’s not already done). For this please click on the settings icon on the monitor using the monitor's touchscreen.
NOTE: Alarm + Function can be used to lock/unlock the monitor's touchscreen, buttons, keys, and rotary wheel:
to lock: Press Alarm + Function keys at the same time until lock icon is displayed at the monitor's screen;
to unlock: Press Alarm + Function keys at the same time until the unlock icon is displayed at the monitor's screen
In the Setup window please select the Network:
And then adjust the values in the window below:
Network Option Description:
MAIN VER - Version on firmware;
CENTRAL - must be BL;
DHCP - must be OFF;
HOST IP - IP Address of the PC where the Surgery Admin application (Listener) installed;
DEVICE IP - IP Address of the BioNet BM7VET Monitor;
SUBNET - Subnet mask of the network, by default is 255.255.255.0;
GATEWAY - It is a start network device IP;
MAC ADDR - It is a media access control address, is a unique identifier.
NOTE: IP address must be different for each device within one network, ie:
HOST IP - PC/Listener IPv4: 192.168.033.100
DEVICE IP - Monitor IPv4: 192.168.033.101
Default Gateway: 192.168.033.001
We hope you found this information useful!
Did you find it helpful?
Sorry we couldn't be helpful. Help us improve this article with your feedback.