How to set up BioNet surgery monitor - BM5VET?
Modified on: Fri, 11 Oct, 2019 at 1:28 PM
IMPORTANT: We are not currently connecting any new customers as we are working to stabilize this integration feature. If you have previously connected and are already using anesthetic monitor integration, please contact our support if you need assistance.
To set up the monitor correctly, please follow the steps below.
Set the correct IP address, Gateway, Subnet Mask and Host IP for the multi-parameter portable veterinary monitor - BM5VET (If it’s not already done). For this please select the icon settingson the monitor using the Touch Wheel and Select Key.
On the Main Menu please scroll to System and press the Trim Knob:
Network Option Description:
MAIN VER - Version on firmware;
CENTRAL - must be ON;
DHCP - must be OFF;
HOST IP - IP Address of the PC where the Surgery Admin application (Listener) is installed;
DEVICE IP - IP Address of the BioNet BM5VET Monitor;
SUBNET - Subnet mask of the network, by default is 255.255.255.0;
GATEWAY - It is a start network device IP;
MAC ADDR - It is a media access control address, is a unique identifier.
NOTE: IP address must be different for each device within one network, ie:
HOST IP - PC/Listener IPv4: 192.168.033.100
DEVICE IP - Monitor IPv4: 192.168.033.101
Default Gateway: 192.168.033.001
We hope you found this information useful!
Did you find it helpful?
Sorry we couldn't be helpful. Help us improve this article with your feedback.