To set up the monitor correctly, please follow the steps below.
Set correct IP address, Gateway, Subnet Mask and Host IP for the multi-parameter portable veterinary monitor - BM3VET Touch, (If it’s not already done). For this please select the settings icon on the monitor using the monitors touchscreen.
NOTE: Alarm + Function can be used to lock/unlock the monitors touchscreen, buttons, keys and rotary wheel:
- to lock: Press Alarm + Function keys at the same time until lock icon is displayed at the monitor's screen;
- to unlock: Press Alarm + Function keys at the same time until unlock icon is displayed at the monitor's screen
In the Setup window please select the Network option:
Then adjust the values as per the instructions below:
Network Option Description:
MAIN VER - Version on firmware;
CENTRAL - must be BL;
DHCP - must be OFF;
HOST IP - IP Address of the PC where the Surgery Admin application (Listener) installed;
DEVICE IP - IP Address of the BioNet BM3VET Touch Monitor;
SUBNET - Subnet mask of the network, by the default is 255.255.255.0;
GATEWAY - It is a start network device IP;
MAC ADDR - It is a media access control address, is a unique identifier.
NOTE: IP address must be different for each devices within one network, ie:
HOST IP - PC/Listener IPv4: 192.168.033.100
DEVICE IP - Monitor IPv4: 192.168.033.101
Default Gateway: 192.168.033.001
For more details please also refer to the official manual: https://www.bionetus.com/bm3vet-touch-screen-veterinary-monitor/