To set up the monitor correctly, please follow the steps below.
Set the correct IP address, Gateway, Subnet Mask and Host IP for the multi-parameter portable veterinary monitor - BM3VET (If it’s not already done). For this please select the settings iconon the monitor using the Touch Wheel and Select Key.
On the Main Menu please scroll to System and then use the Trim Knob to navigate to the next settings window:
Network Option Description:
MAIN VER - Version on firmware;
CENTRAL - must be ON;
HOST IP - IP Address of the PC where the Surgery Admin application (Listener) installed;
DEVICE IP - IP Address of the BioNet BM3VET Monitor;
SUBNET - Subnet mask of the network, by the default is 255.255.255.0;
GATEWAY - It is a start network device IP;
MAC ADDR - It is a media access control address, is a unique identifier.
NOTE: IP address must be different for each devices within one network, ie:
HOST IP - PC/Listener IPv4: 192.168.033.100
DEVICE IP - Monitor IPv4: 192.168.033.101
Default Gateway: 192.168.033.001
Please also set the W-LAN option to OFF:
For more details please also have a look at the official manual: https://www.bionetus.com/patient-monitor/