How to connect Digicare Monitors with Smart Flow's Anesthetic Sheet?
Modified on: Fri, 11 Oct, 2019 at 10:37 AM
IMPORTANT: We are not currently connecting any new customers as we are working to stabilize this integration feature. If you have previously connected and are already using anesthetic monitor integration, please contact our support if you need assistance.
There is an increasing number of multi-parameter anesthetic monitors with the capability to connect directly with Smart Flow through the clinic's internal network. Here are the details for using a Digicare monitor(s) that have already been set up. During the setup process, a small Listener program is installed on a computer in the clinic. The Listener will constantly "listen" to information on the network for data from monitors that are powered on and are on the network.
To begin, you will need to have a patient checked into Smart Flow and setup on the Digicare monitor.
NOTE: It is crucial to follow these steps in the described order. Not doing so may result in connection fail.
From the iPad
2. Once you open the Anesthesia Sheet, please connect a specific monitor. For this tap more actions button (...) > Surgery Monitors (Beta):
3. Select the monitor (e.g. DigiSADental, DigiSAMobile or DigiSASurgery) > Connect:
4. Under Monitoring, a circle icon • will appear, it is yellow while connecting and will turn blue once connected:
NOTE: A red circle will appear if there is an error preventing a connection. Should this happen:
Verify you are connected to the correct monitor.
Verify on the monitor connection. The settings should be set as follows:
IP Address & Port - Should match the IP address that was set up on the Listener.
From the Monitor
5. Connect the Monitor to the Anesthesia Sheet: System > Networking:
Please wait until the Connected status, and click Done:
NOTE: If it does not connect, close it, and try again.
You should now be connected and the vitals should be auto-populating on the Smart Flow Anesthesia Sheet.
We hope you found this information useful!
Did you find it helpful?
Sorry we couldn't be helpful. Help us improve this article with your feedback.